Ulysses HISCALE Pages
Chapter 1 (continued)
Table 1.1 LEMS detector system summary
*Sectors and accumulation times designated with * are associated with the LEMS30 system.
+Sectors and accumulation times designated with + are associated with the LEMS120 system.
Table 1.2 LEFS detector system summary
*Sectors and accumulation times designated with * are associated with the LEFS 150 system.
+Sectors and accumulation times designated with + are associated with the LEFS 60 system.
Table 1.3 Composition aperture summary
|CA 60||W1||C1 D1(C2)(D2)(B1)||0.35-1.0 MeV protons||8||3|
|W2||C2 D1(C4)(D2)(B1)(S1)||1.0-2.0 MeV protons||8||3|
|W3||D2(D3)(C3)(B1)&[C2 or S1]||0.35 - 1.0 MeV/nuc. He||8||3|
|W4||C3 D1 S1(C4)(D3)(B1)||1.0-5.0 MeV/nuc. He||8||3|
|W5||C2 D3(C4)(D4)(B1)||~0.3-1.7 MeV/nuc. CNO||8||3|
|W6||C4 D2 S2(D4)(S3)(B1)||~1.7-10 MeV/nuc. Fe||8||3|
|W7||C2 D4(C4)(B1)||~0.2-1.0 MeV/nuc. Fe||8||3|
|W8||C4 D3 S3(B1)||~1.0-15 MeV/nuc. Fe||8||3|
|Z2||D2(C1)(B1)||≥0.5 MeV Z≥2||4||6|
|Z2A||D3(C1)(B1)||≥10.0 MeV Z≥14||4||6|
|Z3||D5(B1)||≥2.8 MeV Z≥6||4||6|
|Z4||D4||≥9.0 MeV Z≥10||4||6|
(1)Logical anticoincidence required is shown in parentheses.
(2)The passbands given here are provisional and subject to refinement as in-flight calibrations are completed.
Next: Table 1.4
Updated 6/9/09, T. Hunt-Ward